For the cost of £20 ($30 US) it looks and feels like a more expensive meter, but do looks deceive?
The movement is good, you can tell this by moving it from left to right and noting that it is well damped and returns slowly to it zero position, the range selector knobs and switching has a professional feel and click into position.
The manual isn't up to much, as 99% of it is written in Chinese. I am able to decipher that it is an Industry standard 20'000 Ohms/V meter on DC +/- 2.5% , except 2500V @ 4000 Ohm/V. On AC ranges it is 2500 Ohms/V +/- 5% accuracy. DC Current (I) is +/- 2.5% & AC Current +/- 5%.
Ohms measurement +/- 2.5% accuracy.
Batteries required for Ohms ranges are a 9V PP3(6F23), and one C cell, although an AA will fit to get you away.
I was excited like a little boy with a new toy to get the back cover open to give it a good looking over. Turning it upside down onto its face, are 4 sturdy anodised screws that are covered over by plastic plugs in each corner of the case, these easily pop out with small screwdriver forcing them up.
The rear cover then splits away in half, leaving the wires attached from the battery box to the main PCB. I was slightly disappointed by its SMT design, but I accept most things are done this way these days. The odd leaded capacitor and resistor tagged onto the rear of the board did remind me of designs of past years, switching is direct onto the gold plating onto the board. The PCB looks good, there were no signs of any dry joints, however I could not see any fuses for protection, nothing is highlighted in the schematic, what could one really expect for the money? Prehaps I am just used to my ageing AVO 8 with its popout safety breaker.
If you note from the specification the meter does cover 2.5KV! However, I would certainly not like to be on the end of it with the test probes provided (two pairs included), my BIC biro has more plastic on its case than these suicidal leads that are certainly not up to the job! A purchase of a good pair of leads to CAT III spec is a must!
Conclusion so far: Sturdy well made case, Meter movement has the looks and feel of a professional movement costing 10 times more. If you are a novice coming into the hobby and you can't read Chinese don't blame me if you can't understand the manual how to drive it? Anyone else who has been part of the hobby game or the field should have no problem sorting it out. May not be serviceable if you blow or damage parts on the board if you select and use it on the wrong range? Requires a decent set of test leads and probes!
Testing to be concluded further down the Blog.